HANDLING THIS SERVICE MANUAL About This Service Manual n. Scope This Service Manual applies to the Fuji Medical Dry Laser Imager DRYPIX Service Technicians PACS / DR / CR / Digital Forum Drypix Service Manual CARLOS RUEDA Innovacion Tecnologica Hello, I am trying. View and Download FujiFilm DRYPIX service manual online. DRYPIX Printer pdf manual download.
|Published (Last):||22 July 2018|
|PDF File Size:||5.69 Mb|
|ePub File Size:||13.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
Remove the heat development unit.
The DryPix has excellent throughput between and films per servlce making it ideal for modalities such as high speed CT. Page MCT board. Mechanical Jamming, Abnormal Sound 2. FFDM separates out the components of image acquisition, Shutdown the PC for servicing. Tray lock assembly Back side FP7C Remove the filter section cooling fan. Cleaning Roller Grip Release Motor md4 Rotate the worm gear of the cleaning roller grip release motor MD4 in the arrow direction shown in the figure with the hand so that it grips the cleaning roller.
Page When the data transfer is completed, the [File transfer is completed. EPS Remove the sub-scanning unit fixing bracket, and reinstall it to its original position. The measured density steps servjce not exceed the range listed above.
Remove the belt cover. dgypix
Loosen the fixing screws of the magnet. Forums Documents and Servide. Turn ON the power of the equipment, and start the M-Utility. MC Remove the sub-scanning unit fixing bracket. Remove Subscription Confirm Remove. Bellows Remove the suction cup arm. Page 20 MC Remove the tray.
FUJIFILM DRYPIX SERVICE MANUAL Pdf Download.
It accommodates all common film sizes, and has an optional second film drawer and top-mounted 4-bin film sorter. If the equipment mounts film tray 2, loosen the screws securing the idle gear of the middle conveyor unit.
Page of Go. However, there is no need to pull out 400 rail pushed into the equipment. Page 3 The screws painted in yellow must be adjusted after the installation of parts. Remove the cooling fan assembly.
service manual – Fujifilm DryPix | MedWrench
Output the grid pattern using the M-Utility. Film Positioning Motor servics md5 Remove the upper conveyor unit.
Remove the SUS belt. There may be slight variations in the characteristics of the film between batches.
Add to My Bench. Page 24 Reinstalling Procedure Reinstall in the reverse order of removal.
Dear sir, I am a service engineer free lance. Got it, continue to print. When completed, [File Copy Finish!! Some variation from the published values may be caused by slight differences between the measurements of the srvice printer densitometer and the external densitometer used at the site. Install the inside belt first and then the top and bottom belts. Not only does each mammographic x-ray unit model have Comments to this Manuals Your Name.
Remove the suction cup arm.
MC If the measured values are not the specified values, loosen the screws securing the sub-scanning unit. Page 82 Secure the gear while pressing the flange of the idle gear against the gears in the arrow direction shown below.
The DryPix is a dry laser imager which combines the high quality and performance of Fuji’s centralized imagers, with an extremely compact design and convenient volt power, required for decentralized imaging applications.
Don’t have an account? Rotate the worm gear of the cleaning roller grip release motor MD4 in the arrow direction shown in the figure with the hand so that it grips the cleaning roller.
Page 89 MC If the measured values are not the specified values, loosen the screws securing the sub-scanning unit. When reinstalling the adiabatic cover, do not sandwich the thermistor cable. You can adjust your Community Drpix in Settings. Page 52 Push down the plate, and with an approximately 50 mm space between the plate and stopper, push the suction cup against the film. Two minutes later, check that the space is still 50 mm. To print the manual completely, please, download it.