We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||11 September 2007|
|PDF File Size:||6.87 Mb|
|ePub File Size:||8.38 Mb|
|Price:||Free* [*Free Regsitration Required]|
Troubleshooting Alarms and Troubleshooting Dialysate Screen Fresenijs Deaeration Motor Brushes Chapter 3—Setting Treatment Parameters New Treatment Key When the K is first turned on in preparation for daily operation after a long power downall treatment parameters revert to their default settings.
Page 64 The operator can start or stop the pure UF option at any time. Fill Program—Occurs when water level in the air separation chamber gets too low.
Section fresenkus — Hydraulic Leaks Replacing The 9-volt Battery If unable to reset the alarm, call your local qualified technician. Test for residual disinfectant prior to starting treatment following a chemical disinfection. Manufacturer’s Emc Declaration Fail Neg Stabilize This will minimize the possibility of the segment kinking during pump operation.
Fresenius Medical Care K Manuals
Arterial, Venous And Transmembrane Pressure The range of the display is different depending on the location of the chamber. Page Start key a second time and the pump segment will be released from the pump head. Improper bloodline connections or needle dislodgements can result in excessive blood loss, serious frdsenius, and death. A button refers to pressure-sensitive area located in the treatment screens that is used to set treatment parameters.
During heat disinfection, the RO unit, if used, can be disconnected from the K once the water temperature reaches 80 C and the recirculation cycle starts.
Acid Cleaning Acid Cleaning Acid Cleaning The K should undergo an acid frseenius daily when using bicarbonate concentrates during dialysis. Observe the system for fluid leaks and discontinue treatment fresenijs you are unable to correct any fluid leak quickly. Section 14 – Uf ultrafiltration Pump Problems The table consists of four columns: Blood Recirculation Procedure An audible alarm will sound every two minutes to alert the operator that blood is sensed while the TX Clock is paused.
Heater is off during pre-rinse. Daily Preparation For Treatment Page 29 is depressed.
Safety class I Degree: The up or down arrow keys on the control panel may be used to change the time. The UF Time button acts as a countdown timer displaying the remaining time ultrafiltration will be performed.
Fresenius Medical Care 2008K Operator’s Manual
Now its possible because of problems and lack of response that it could take a few years. Home Screen Settings Preparing The Extracorporeal Blood Circuit The red lines on the machine are guides for arterial bloodline from patient to dialyzer.
A minor blood leak has been detected by the blood leak detector. Fail Long Pulse” Or “wd: May also indicates a hardware malfunction. Assembly, installation, adjustment, or repair is to be performed only by persons authorized by the facility medical director or by Fresenius USA.
Surprised non of you “F-ers” ever volunteered to be the lean facilitator for your clinics since it was suppossed to be a grass roots effort.